Cos ϕ vs. power factor λ

Practice

This second technical report on the subject of reactive power and in particular the distortion reactive power supplements the first contribution No. 05 (cosphi vs. power factor theory). In this report the voltage and current of an light bulb is measured with a power analyzer (PQ-Box 200) in an online measurement, which is controlled by a phase angle control. This measurement example is used to illustrate the different types of reactive power.

The example shows how to use a power analyzer to measure the voltage, current and power readings once on the mains side at the socket and once directly at the load, in this case a light bulb. With these measured values, one can then calculate and analyze the distortion reactive power.

In the online oscilloscope image, the waveforms of current and voltage are visible. At first, only the
voltages are displayed. A sine wave of the socket can be seen, which has a clean shape, but is slightly
flattened. This is typical for office buildings where a large number of single-phase power supplies are used.
The line voltage has an RMS value of nearly 230V UL1. When the phase angle control is set to maximum brightness, the RMS value of the current is 1.33A

The voltage is now cut via the phase angle control until an effective current value of 1A is reached. The current is identical at every point of the circuit and is also 1A RMS at the light bulb.
The voltage at the light bulb was reduced to an RMS value of 135V. Since the light bulb is an ohmic load, the current follows the applied voltage 1:1. In the measurement results, the voltage at the socket is shown in green and the voltage given via the phase angle control of the light bulb is shown in blue.

In the measurement, a voltage of 135 volts and a current of 1 ampere were measured at the light bulb. On the mains side, 230 volts and 1 ampere were measured. So the question arises which power measurement values result on the mains side as well as at the consumer? Which power is calculated as active power?
The use of phase angle control results in a shift of the fundamental oscillation voltage to the fundamental oscillation current. This gives us a phase angle phi of the fundamental shift. In this case the current is 40 degrees lagging to the voltage. The phase angle on phases L2, L3 (on the light bulb) is almost zero, which corresponds to a cosine phi of exactly 1. It should be noted that the light bulb remains an ohmic load.

Figure No. 5 shows all power measurement values of the measurement inputs of the PQ-Box 200. All measurement values of phase L1 refer to the power supply (socket), the measurement values L2 and L3 were determined directly at the light bulb.

This results in the following measured power values:

1. Measured values at the light bulb (L2, L3)
The active power corresponds to the apparent power. The cosphi as well as the power factor PF are at the value 1, which corresponds to a purely resistive load.
2. Measured values at the mains connection (before phase angle control; L1)
Due to the phase shift from fundamental current to voltage with approx. 40°, there is a fundamental
reactive power of 114Var. This reactive power alone cannot explain the difference between the
apparent power (232VA) and the active power (140W). The power factor that results from the division of the apparent and active power is 0.6. The cosphi that is calculated from the phase angle phi results in a value of 0.77.

If the active power at the input of the circuit is calculated with the known formula P = U x I x Cos(Phi) the measured values will be incorrect. In our case an active power of 177W. In order for the formula to calculate a correct active power, it must be ensured that only the 50Hz section of the current is used for the electricity. This corresponds to 1A.

An analysis of the current spectrum shows an RMS value of 1A, which is the sum of the fundamental and all current harmonics. If the cursor is positioned on 50 Hertz, the WinPQ mobil software shows a value of 770mA and, for example, on 150 Hertz a value of 118mA. A correct calculation of the active power requires the use of 230 volts multiplied by the correct current, which is limited to the 50 hertz component of 0.770A and the cosine of the angle Phi of 0.77. The result now corresponds to the value converted by the light
bulb. The meter already calculates this correctly, resulting in an active power of 139 watts on the mains side. The light bulb currently converts 136 watts. The difference of a few watts is due to losses of the dimmer.

In this circuit, the harmonic reactive power is not negligible. This can be seen in graph 9 (marked D).

In this measurement, the collective total reactive power is composed of the fundamental reactive power of 114Var and a distortion reactive power (harmonic reactive power) of 147Var. These summed squarely give a collective total reactive power of 186Var. The active power is 138W. The fundamental displacement reactive power of 114Var is at right angles to the active power (green). The square sum of the two reactive powers gives the total reactive power for phase L1 (red) of 186var. The active power and total reactive power summed squared gives 232VA (blue), which again corresponds to U times I. It must be noted that in electrical installations, if the current is not sinusoidal, there is a distortion reactive power and this must be included in the calculation of the powers.

Author
Jürgen Blum, Product Manager Power Quality Mobil

Contact

News from the product group

PQMobil - News

General Knowledge

Voltage Monitoring

Find out everything you need to know about voltage monitoring in this article. What is voltage monitoring, why is voltage monitoring essential for both energy supply companies and industrial companies and how is it implemented in compliance with standards?

General Knowledge

What Is Load Profile Measurement/Power Measurement?

This article explains the definition and difference between load profile measurement (RLM), power measurement and measuring the standard load profile. These are important methods for accurately recording energy consumption in commercial, large industrial plant as well as in the public energy grid.

Training Video

»PQMobil« Training Video 4: Create and Configure PQ Reports With Software WinPQ mobil

This video explains everything about creating and configuring power quality reports with »WinPQ mobil«.

Training Video

»PQMobil« Training Video 3: Usability & Functions of the PQ-Box App

This video gives you an introduction to the usability and functions of the »PQ-Box App« for our portable power quality analysers.

Training Video

In this short introductory video, we present to you how to navigate and operate our portable power quality analysers »PQ-Box 150«, »PQ-Box 200« & »PQ-Box 300« of the PQ-Box family.

Training Video

»PQMobil« Training Video 1: Getting Started with the PQ-Box

This video shows in detail how to connect the »PQ-Box« and record measurements. The »WinPQ mobil« software is then used to visualize and analyse the measurement data and create a report.

General Knowledge

Residual current

In this article, you will learn about residual current measurement & residual current monitoring and why it is important to measure residual current.

General Knowledge

Power Quality

Good power quality is characterized by the fact that the mains voltage actually arriving at the consumer matches the mains voltage promised by the utility company.

General Knowledge

Reactive power – definition, calculation and measurement

The key role of reactive power in electrical engineering: definition, calculation, measurement and the differences to active power and apparent power. Find out how reactive power influences the efficiency of electricity grids and shapes energy transmission.

General Knowledge

What is apparent power and how is it calculated?

This article addresses the question of what apparent power is and how to calculate it. An understanding of apparent power is crucial for the correct dimensioning of inverters and therefore for the optimum efficiency and performance of photovoltaic systems.

Info letter

Information Letter No. 24: Integration guide for the metering of a power plant according to the Requirements for Generators (2016/631) RfG

Integration guide for the measurement of a power plant (EZA) according to VDE-AR 4110 and VDE-AR 4120.

Info letter

Info Letter No. 23: Direction of harmonics Sources of harmonics in the power system

Le sens du flux de puissance des harmoniques est déterminé par le signe de la puissance activ

Info letter

Info Letter No. 22: Definition of power measurements according to the standards DIN 40110-2 and IEEE 1459

There is a wide range of electronic measuring devices for the digital measurement of power...

Info letter

Info Letter No. 20: Capacitance of conductors

In the conductors of electrical power supplies, a distinction is made between the operating capacitance...

Info letter

Info Letter No. 19: The evaluation of power quality and detection of interference in medium voltage network

For various reasons, interest in the continuous monitoring of network quality in medium-voltage...

Info letter

Info Letter No. 8: Apparent power and reactive power in three-phase networks

The range of electromechanical and electronic measuring devices in analog - and digital technology...

Info letter

Info Letter No. 5: Zero sequence voltage in three-phase networks

With balanced network operation and inequality of the impedances in the consumer circuit, the phase...

Info letter

Info Letter No. 4: Summation of Alternating Currents

For the parallel connection of power sources, the principle of superposition applies, when all resources...

Info letter

Info Letter No. 2: Measurement Transformers in three-phase networks (Part 2)

In a symmetrically loaded three-wire three-phase network, a two-pole insulated voltage transformer...

Info letter

Info Letter No. 1: Measurement Transformers in three-phase networks (Part 1)

The direct connection of measuring devices into the network is limited for safety reasons to values...

Webinar

»PQMobil« Webinar: Grid Analysis in Public & Industrial Power Grids with A. Eberle PQ-Boxes

Webinar recording from 2024-02-22: »PQMobil - Grid Analysis in Public & Industrial Power Grids with A. Eberle PQ-Boxes«.

Special Publication

Ferrari´s Meter vs. Electronic Meter

This technical report analyzes a power measurement at a large industrial customer. The customer records very high currents and fifth harmonic voltages during production time, which leads to poor voltage quality.

Special Publication

Angle determination of voltage and current harmonics in practice

Today's devices and equipment such as switched-mode power supplies, frequency converters, regulated drives, charging devices for e-mobiles or LED lighting work internally with high clock frequencies in order to regulate power efficiently. These can lead to both conducted and field-bound (coupled) influences in the energy grid. In this technical report, we explain how you can detect these sources of interference in the grid using suitable measuring technology.

General Knowledge

Today's devices and equipment such as switched-mode power supplies, frequency converters, controlled drives, charging devices for e-mobiles or LED lighting work internally with high clock frequencies in order to regulate power efficiently. These can lead to line-borne as well as field-borne (coupled) interference in the power grid. In this technical report, we explain how you can detect these sources of interference in the grid using suitable measurement technology.

General Knowledge

Power Quality Analysis: Tips and Tricks II

Useful tips & tricks for handling Power Quality measuring devices and typical user errors - now in our two-part article! This article provides further insights and practical tips and application examples for power quality analysis.